MZC-310S

Index: WMGBMZC310

Loop Meter

Description

Technical Specification

Files

- DescriptionThe MZC310S is a professional portable meter for testing electrical installations with over-current circuit breakers. The instrument measures short-circuit L-PE, L-N and L-L loop and prospective short-circuit loop current. A measurements can be done either with low testing current (up to 42 A) with 2-pole method or high testing current (up to 280 A) with 4-pole method that enables measurements with very high accuracy and resolution.The results can be stored in the internal memory and send to a computer via serial interface.
- Technical Specification
**Measurement of very low short circuit loop impedances (with resolution 0,1 mΩ) with a current of 150 A at 230 V; maximum 280 A at 440 V or measurement with a current of 23 A at 230 V; maximum 42 A at 440 V.**- measurements in installations with rated voltages between: 220/380 V and 230/400 V and frequencies 45...65 Hz,
- ability to perform measurements in short circuit system: phase-phase, phase-protective, phase-neutral,
- differentiation between the phase voltage and the inter-phase voltage while calculating the shortcircuit current,
- Ability to change the length of test leads (measurement 23/42 A),
- four-pole method, test leads do not require calibration (measurement 150/280 A).

**Touch voltage and touch shock voltage measurement (with resistor 1 kΩ).****AC voltage measurement in range 0...440 V.****Memory for 999 measurement results with an ability to transfer the data to a PC.****Meter meets the requirements of the standard EN 61557.****Electric security:**- type of insulation: double, according to EN 61010 - 1 and IEC 61557
- measurement category: CAT IV 300 V acc. to EN 61010 - 1
- protection class acc. to EN 60529: IP20

**Other technical data:**- power supply alkaline batteries LR14 (C) (5 pcs.)
- resistor limiting the current:
- for measurement 4p: 1,5 Ω
- for measurement 2p: 10 Ω
- number of short circuit loop measurements (alkaline batteries): min. 2000 (4/min.)
- temperature coefficient: ±0,1% of measured value /°C

**Rated operational conditions:**- operating temperature: 0...+40°C

Range Resolution Accuracy 0...440 V 1 V ±(2% m.v. + 2 digits) Range Resolution Accuracy 45,0...65,0 Hz 0,1 Hz ±(0,1% m.v. + 1 digits) Range Resolution Accuracy 0...199,9 mΩ 0,1 mΩ ±(2% m.v. + 2 mΩ) 200...1999 mΩ 1 mΩ ±(2% m.v. + 2 mΩ) Range Resolution Accuracy 0...199,9 mΩ 0,1 mΩ ±(2% m.v. + 2 mΩ) impedance reading for a particular measurement 200...1999 mΩ 1 mΩ ±(2% m.v. + 2 mΩ) impedance reading for a particular measurement Range Resolution Accuracy 115,0...199,9 A 0,1 A accuracy of the current indication computed, respectively, with the use of resistance measurement 200...1999 A 1 A accuracy of the current indication computed, respectively, with the use of resistance measurement 2,00...19,99 kA 0,01 kA accuracy of the current indication computed, respectively, with the use of resistance measurement 20,0...199,9 kA 0,1 kA accuracy of the current indication computed, respectively, with the use of resistance measurement 200 kA...* 1 kA accuracy of the current indication computed, respectively, with the use of resistance measurement Range Resolution Accuracy 0...100 V 1 V ±(10% m.v. + 2 digits) Range Resolution Accuracy 0,00...19,99 Ω 0,01 Ω ±(2% m.v. + 3 digits) 20,0...199,9 Ω 0,1 Ω ±(3% m.v. + 3 digits) Range Resolution Accuracy 0,00...19,99 Ω 0,01 Ω ±(2% m.v. + 3 digits) impedance reading for a particular measurement 20,0...199,9 Ω 0,1 Ω ±(2% m.v. + 3 digits) impedance reading for a particular measurement Range Resolution Accuracy 1,150...1,999 A 0,001 A accuracy of the current indication computed, respectively, with the use of resistance measurement 2,00...19,99 A 0,01 A accuracy of the current indication computed, respectively, with the use of resistance measurement 20,0...199,9 A 0,1 A accuracy of the current indication computed, respectively, with the use of resistance measurement 200...1999 A 1 A accuracy of the current indication computed, respectively, with the use of resistance measurement 2,00...19,99 kA 0,01 kA accuracy of the current indication computed, respectively, with the use of resistance measurement 20,0...40,0 kA 0,1 kA accuracy of the current indication computed, respectively, with the use of resistance measurement „m.v.” - measured value

- Files