MPI-540 + EVSE-01
SONEL MPI-540 + SONEL EVSE-01 SET
Sonel MPI-540 | Much more than a multifunctional meter
The Sonel MPI-540 multifunctional meter is intended for testing home and industrial electrical installations. The instrument can be used to carry out measurements, the results of which determine the safety status of the installation. In addition, the user is able to record the parameters of the electrical networks located at these sites. This makes it possible to control the quality of electricity, and to measure electric shock protection parameters, using a single universal device.
Significant automation of measurements is made possible by the function of testing residual current circuit breakers in Auto mode, as well as measurement sequences pre-programmed by the manufacturer (so-called self-tests), which can also be extended with own sequences. Automatic measurement of insulation resistance of 3-, 4- and 5-core cables will be carried out with the Sonel AutoISO-1000C adapter.
Sonel EVSE-01 | Quick and full diagnostics of charging stations
Sonel EVSE-01 adapter allows to perform comprehensive measurements of electric vehicle charging stations - quickly and in accordance with applicable regulations. Simulating the charging cable (proximity pilot line - PP) and vehicle connection status (control pilot line - CP), it will bring the station into different operating states. This will enable measurements in the field of electric shock protection: ZS fault loop impedance, RISO insulation resistance and checking the parameters of RCD residual current devices.
To facilitate diagnostics, one of the Sonel EVSE-01 sockets is provided with pulse width modulation signal (PWM).
The Sonel EVSE-01 adapter enables measurements of AC electric vehicle charging stations with type 2 connector. Tests for 1-phase and 3-phase stations are available - both with and without ventilation.
- Technical Specification
Measurement functions Measurement range Display range Resolution Accuracy Fault loop impedance Fault loop ZL-PE, ZL-N, ZL-L 0.13 Ω…1999.9 Ω acc. to IEC 61557 0.00 Ω…1999 Ω from 0.001 Ω ±(5% m.v. + 30 digits) Fault loop ZL-PE in RCD mode from 0.50 Ω…1999 Ω acc. to IEC 61557 0.00 Ω…1999 Ω from 0.01 Ω from ±(6% m.v. + 5 digits) Measurements of RCD parameters RCD tripping test and measurement of tripping time tA measuring current 0.5 IΔn, 1 IΔn, 2 IΔn, 5 IΔn general and short-time delay RCD 0 ms…300 ms 0 ms…300 ms 1 ms from ±(2% m.v. + 2 digits) selective RCD 0 ms…500 ms 0 ms…500 ms 1 ms from ±(2% m.v. + 2 digits) Measurement of RCD tripping current IA measuring current 0 0.2 IΔn…2.0 IΔn for sinusoidal residual current (AC type) 3.3 mA…1000 mA 3.3 mA…1000 mA from 0.1 mA ±5% IΔn for unidirectional residual current and unidirectional with the 6 mA DC bias (type A) 3.5 mA…700 mA 3.5 mA…700 mA from 0.1 mA ±10% IΔn for direct residual current (type B) 2.0 mA…1000 mA 2.0 mA…1000 mA from 0.1 mA ±10% IΔn Earth resistance 3- and 4-pole method from 0.50 Ω…1.99 kΩ acc. to IEC 61557-5 0.00 Ω…1,99 kΩ from 0.01 Ω from ±(2% m.v. + 3 digits) 3-pole + clamp method 0.00 Ω…1.99 kΩ 0.00 Ω…1.99 kΩ from 0.01 Ω from ±(2% m.v. + 4 digits) 2-clamp method 0.00 Ω…99.9 kΩ 0.00 Ω…99.9 kΩ from 0.01 Ω from ±(10% m.v. + 4 digits) Resistance-to-earth 0.0 Ω m…99.9 kΩm 0.0 Ωm…99.9 kΩm from 0.1 Ωm Depending on accuracy of RE measurement Insulation resistance Measuring voltage 50 V 50 kΩ…250 MΩ acc. to IEC 61557-2 0 kΩ…250 MΩ from 1 kΩ from ±(3% m.v. + 8 digits) Measuring voltage 100 V 100 kΩ…500 MΩ acc. to IEC 61557-2 0 kΩ…500 MΩ from 1 kΩ from ±(3% m.v. + 8 digits) Measuring voltage 250 V 250 kΩ…999 MΩ acc. to IEC 61557-2 0 kΩ…999 MΩ from 1 kΩ from ±(3% m.v. + 8 digits) Measuring voltage 500 V 500 kΩ…2.00 GΩ acc. to IEC 61557-2 0 kΩ…2.00 GΩ from 1 kΩ from ±(3% m.v. + 8 digits) Measuring voltage 1000 V 1000 kΩ…9.99 GΩ acc. to IEC 61557-2 0 kΩ…9.99 GΩ from 1 kΩ from ±(3% m.v. + 8 digits) Resistance of protective conductors and equipotential bondings Measurement of resistance of protective conductors and equipotential bondings with ±200 mA current 0.12 Ω…400 Ω acc. to IEC 61557-4 0.00 Ω…400 Ω from 0.01 Ω ±(2% m.v. + 3 digits) Measurement of resistance with low current 0.0 Ω…1999 Ω 0.0 Ω…1999 Ω from 0.1 Ω ±(3% m.v. + 3 digits) Light intensity Measurement in luxes (lx) 0 lx…399.9 klx 0 lx…399.9 klx from 0.001 lx from ±(2% m.v. + 5 digits) Measurement in feet-candles (fc) 0 fc…39.99 kfc 0 fc…39.99 kfc from 0.001 fc from ±(2% m.v. + 5 digits) Phase sequence indication in the same direction (correct), opposite direction (incorrect), UL-L voltage: 95 V…500 V (45 Hz…65 Hz) Parameter Measuring range Max. resolution Accuracy Alternating voltage (TRMS) 0.0…500 V 0.01% Unom ±0,5% Unom Alternating current (TRMS) depending on clamp* 0.01% Inom ±2% m.v. if m.v. ≥ 10% Inom, ±2% Inom if m.v. < 10% Inom , (error does not account for clamp error) Frequency 40.00…70.00 Hz 0.01 Hz ±0.05 Hz Active, reactive, apparent and distortion power depending on configuration (transducers, clamps) 4 significant digits depending on configuration (transducers, clamps) Active, reactive and apparent energy secondo la configurazione (sensori, pinze) 4 significant digits as power error cosφ and power factor (PF) 0.00…1.00 0.01 ±0.03 Harmonics Voltage as for alternating voltage True RMS as for alternating voltage True RMS ±5% m.v. if m.v. ≥ 3% Unom, ±0,15% Unom if m.v. < 3% Unom Current as for alternating current True RMS as for alternating current True RMS ±0,5% m.v. if m.v. ≥ 10% Unom, ±0,5% Unom jeśli m.v. < 10% Unom THD Voltage 0.0…100.0% (relative to RMS value) 0.1% ±5% Current 0.0…100.0% (relative to RMS value) 0.1% ±5% Unbalance factor 0.0…10.0% 0.1% ±0.15% (absolute error) C-4A C-5A C-6A C-7A F-1A F-2A F-3A Rated current 1000 A AC 1000 A AC
1400 A DC
10 A AC 100 A AC 3000 A AC 3000 A AC 3000 A AC Frequency 30 Hz…10 kHz DC…5 kHz 40 Hz…10 kHz 40 Hz…1 kHz 40 Hz…10 kHz 40 Hz…10 kHz 40 Hz…10 kHz Max. diameter of measured conductor 52 mm 39 mm 20 mm 24 mm 380 mm 250 mm 140 mm Minimum accuracy ≤0,5% ≤1,5% ≤1% 0,5% 1% 1% 1% Battery power - √ - - - - - Lead length 2.2 m 2.2 m 2.2 m 3 m 2.5 m 2.5 m 2.5 m Measurement category IV 300 V IV 300 V IV 300 V III 300 V IV 600 V IV 600 V IV 600 V Ingress protection IP40 IP40 IP40 IP40 IP67 IP67 IP67 MPI-540-PV MPI-540 MPI-535 MPI-530-IT MPI-530 MPI-525 MPI-520 MPI-507 MPI-506 MPI-502 auto measurements √ - - - automatic threephase measurement via multiplug √ - - - visual inspection √ - - - fault loop ZL-PE, ZL-N parameters measurement √ √ √ √ 6 mA RCD test √ - - - RCD test AC, A, F, B, B+, EV AC, A, F, B, B+ AC, A AC, A insulation resistance RISO measurement √ √ √ - measurements report √ √ √ √
- Sonel Analysis 4changelog
- Added new nominal voltage Un 346/600 V.
- Expanded micro-installations recommendation report.
- Fixed data exclusion in standards reports for PQM-702(x)/703/710/711 (requires at least firmware 1.55) and PQM-700 (requires at least firmware 1.18) recorded data.
- Fixed possibility to enable detection of waveshape variations and phase jumps events in half-period recording configurations for PQM-702(x)/703/710/711 (requires at least firmware 1.55).
- Improved recovery data mechanism.
- Minor bug fixes.
- added support for 4-quadrant energy meters for MPI-540 and MPI-540-PV recorded data (requires at least firmware 3.21);
- added new nominal voltage Un 462/800 V;
- increased micro-installations recommendation report.
- improved standard reports export;
- minor bug fixes.
- Added support for the PQM-702/703 analyzers without the OR-1 interface.
- Translation fixes in Ukrainian language.
- Added Ecuadorian standards for PQM-702(x)/703/710/711.
- Increased edit ability of the reports header.
- Implemented A3:2019 amendment to EN 50160 standard (requires restoring default values for "EN 50160:2010 + A3:2019 low voltage" standard in Program settings module).
- Fixed problem with reading NEC220.87 data.
- Improved user interface.
- Minor bug fixes.
- Sonel Readerchangelog
Changes in version 18.104.22.168 compared to the previous one:
- communication with MIC-2511 updated.
- Firmware MPI-540changelog
Changes in version 3.23.02 compared to the previous one:
Improved memory structure.
Improved lighting measurements.
Improved performance of the battery level indicator.
Fixed minor bugs in the interface.